Turn the meter on by pressing the button,
all segments will be displayed. When the
display shows ”Add”, ”C.1” with ”Press”
blinking, the meter is in a position.
• Fill the cuvette to the 10 mL line at the
cuvette with unreacted sample and replace
the cap. Place the cuvette into the meter
and close the meter’s cap.
• Press the button. When the display shows
” with ”
” blinking the
meter is zeroed.
Note: Any chlorine present in the sample will
interfere with the reading. To take away the
chlorine interference add one drop of
Chlorine Remover to the
• Cast off the cuvette, open it and the usage of a
1 mL syringe carefully add exactly 1.00 mL
of Alkalinity Reagent to the sample. Replace
the cap and gently invert 5 times. Place the
cuvette back into the meter.
: Pay attention to not spill reagent in a different way
full color development is also inhibited.
• Press the button. The instrument instantly
displays the concentration of alkalinity in
ppm of CaCO3
. Alkalinity conversion:
1 ppm CaCO
= 0.02 meq/L
= 0.056 dkH
The meter routinely turns off after 10
Alkalinity Reagents for Salt Water Checker – Reagents for 50 Tests
ONLY works with the Hanna HI755 Alkalinity Checker for SALT WATER